Continuity Measurement Continuous testing can be carried out by use of the test button lockdown feature. A selectable buzzer gives instantaneous indication of continuity. Null facility eliminates the test lead resistance from the results, the nulled value is retained even if the instrument is switched off. Live circuit warnings are given by a flashing LED, buzzer and indication on the display.
Insulation Measurement Three selectable test voltages 250V, 500V and 1000V. An auto-discharge function ensures that circuits are not hazardous after testing. A red LED gives warning of high voltage output during testing and discharging of the circuit. In case of connecting to a live circuit, a live circuit warning is given by flashing LED, buzzer and indication on the display.
Loop Impedance Measurement A patented (ATT) low current loop impedance test enables high accuracy loop measurements and quick testing without tripping RCD's.
A high current alternative is selectable for even higher accuracy and instantaneous results. The subsequent test will default to the low current test, this saves any inadvertent tripping of the RCD. The KEW 6016 allows also for phase to phase loop tests.
The Prospective Short Circuit Current (PSC) and Prospective Fault Current (PFC) are automatically calculated and shown on the display. As loop testing, the function has low and high test current options with the default to low current to avoid inadvertent tripping of RCDs.
RCD Measurement The KEW 6016 has a comprehensive RCD test feature for RCD type AC (Alternative Currents), RCD type A (Pulsating Direct Currents), General and Selective (delayed).
Measures at 1/2x, 1x, 5x of nominal RCD current. It also has Ramp Test and Auto test where all results are shown on one screen. Touch voltage limit can be selected for 25V or 50V depending on application.
Earth Measurement Using the classical Volt-Amper method with two auxiliary earth spikes and without external power source. All test leads and spikes are supplied as standard accessories.
Phase rotation KEW 6016 can check the phase rotation of three phase lines with clear indication of the sequence on the display.
Voltage Measurement In addiction to the voltage measurement, this function gives also the Frequency of the voltage under test.
Nominal Test Current at 0O External Loop: Magnitude/ Duration (*2)
20O: 6A/20ms 200O: 2A/20ms 2000O: 15mA/500ms
L-N: 6A/60ms N-PE: 10mA/approx. 5s
20/200/2000O (Auto-ranging) (L-N < 20O)
±3%rdg+6dgt (*3) ±3%rdg+8dgt(*4)
±3%rdg+4dgt (*3) ±3%rdg+8dgt(*4)
PSC (L-N / L-L) / PFC (L-PE)
100 - 500V 50/60Hz
100 - 260V 50/60Hz
Nominal Test Current at 0O External Loop: Magnitude/ Duration (*5)
6A/20ms 2A/20ms 15mA/500ms
L-N: 6A/60ms N-PE: 10mA/approx. 5s
PSC / PFC accuracy is derived from measured loop impedance specification and measured voltage specification
Trip current setting
10/30/ 100/300/ 500mA
Depending on the accuracy at each function. Measurement sequence: X1/2 0°?X1/2 180° ?X1 0°?X1 180° ?X5 0°?X5 180° Measurements with x5 are not carried out for RCDs with nominal current of 100mA or more.
Trip current Duration
G: 550ms S: 1000ms
Goes up by 10% from 20% to 110% G:300ms/ S:500ms X10 times
Correct phase sequence: are displayed "1.2.3" and cycle mark Reversed phase sequence: are displayed "3.2.1" and cycle mark
25 - 500V, 45 - 65Hz
25 - 500V
45 - 65Hz
IEC 61010-1 CAT III 300V (500V L to L) Pollution degree 2, IEC 61010-031, IEC 61557-1, 2, 3, 4, 5, 6, 7, 10, IEC 60529 (IP40), IEC 61326 (EMC)
LR6 X 8
136(L) X 235(W) X 114(D)mm
1350g (including batteries.)
7188A (Distribution board fuse test lead set) 7196A (Test leads with remote control switch) 7218A (Power cord [EU]) (*6) 7228A (Earth resistance test leads) 8032 (Auxiliary earth spikes) 8212-USB (USB adaptor with KEW Report (Software)) 9142 (Carrying case) 9121 (Shoulder strap) Cord case, Buckle, Battery, Instruction manual
OBIAT Electronics Pty Ltd | Test & Measurement Equipment
OBIAT Electronics has been established as an importer, distributor, wholesaler and retailer of quality Test and Measuring Instruments for over fifteen years.
Our competent staff collectively have ...