Temperature compensation is accomplished with a built-in sensor.
Application includes water treatment, cooling tower and water monitoring.
Three models are available for specific measuring range.
ST3254.1 Range 0-10mS
ST3254.2 Range 0-100mS
ST3254.3 Range 0-1000mS
ST3214.5 Range 0-200mS
PRINCIPLE OF OPERATION
When the electrodeless conductivity sensor is immersed in the sample to be measured, a conductive loop is created through the two toroidally wound coils. An Alternating current is applied to one of the coils which induces a current in the conductive loop.
The second coil is udes to measure the conductivity which is proportional to the induced current in the solution.
The advantages of the electrodeless method are more apparent in measurement applications in which electrodes contamination and polarization of a conventional conductivity system can lead to erroneous readings.
Each probe contains:
- Two measuring toroidal coils
- Temperature sensor
- 4-20mA current loop amplifier