All of the components exposed to the high temperature gases are ceramic or platinum, for optimum corrosion resistance. It is of robust construction, with an alumina ceramic sheath to protect the sensing element.
The probe can be installed anywhere in the kiln or furnace, provided that the temperature at the tip is between 700°C and 1700°C. The probe will measure the oxygen concentration at the probe tip.
The port for installation should be sealed sufficiently to avoid atmospheric air from flowing inwards and causing a local variation in the oxygen concentration.
The probe can be used at any orientation for temperatures up to 1100°C, but should hang vertically if used at higher temperatures.
The probe head should be maintained at a temperature less than 60°C.
Clean respirable reference air at a rate of 10 cm3/min must be supplied to the connection provided on the probe head.
For further information or a price quotation, please call or use the IndustrySearch enquiry box for a direct reply